Electromagnetic Simulation Helps Design Next Generation of MRI Equipment
Fri, 12/05/2008 - 7:03am
The Project: Gain a better understanding of how the RF transceiver interacts with the nuclei in MRI systems in order to be able to design the next generation.
The Solution: Use electromagnetic simulation software to demonstrate how these interactions occur.
Dr. Christopher M. Collins is an associate professor of radiology at Pennsylvania State University College of Medicine. He received his Ph.D. in Bioengineering from the University of Pennsylvania in 1999 and since then has worked at Penn State's College of Medicine to improve the efficacy and safety of MRI through improved understanding of interactions between human tissues and electromagnetic fields. Dr. Collins can be reached at 717-531-7402 or firstname.lastname@example.org.
A new generation of high definition magnetic resonance imaging (MRI) machines is enabling researchers to gain a far more accurate understanding of the metabolic pathways of the human brain and how these pathways are affected by disease. This new generation is generally known as 7-Tesla, which refers to the strength of the static magnetic fields generated by the machines. These fields are used to cause the nuclei in the body to wobble, or precess, so that a radio frequency (RF) transceiver can detect their movement and generate images. This new generation of machines operates at a much higher frequency than conventional MRI, requiring a better understanding of how the RF transceiver interacts with the nuclei. Electromagnetic simulations were critical in demonstrating how these interactions occur, which has provided important insight in the design of the new generation of machines.
How MRI Works
Due to the nature of this nuclear precession, it is the circularly polarized components of the RF fields that are important in communicating with the nuclei. Until fairly recently, there was disagreement about which circularly polarized field components were important in communicating with the nuclei. There was some evidence and theory indicating that the RF magnetic field component rotating in the same direction of nuclear precession was important during excitation and the component that rotated in the opposite direction of nuclear precession was important during signal reception. To many working in the field of MRI, this seemed counter-intuitive. They were convinced that the component that rotated in the same direction as nuclear precession should be the active component during both excitation and reception.
Moving to the Next GenerationThis discussion was of only theoretical interest during the first few generations of MRI equipment. But it became much more important as MRI moved to higher static magnetic fields in order to increase the signal-to-noise-ratio (SNR) and improve the resolution of the resulting images. The latest 7-Tesla MRI machines have a field strength which is 140,000 times stronger than the earth's magnetic field. This field is typically generated by a magnet that holds between 400 and 500 kilometers of superconducting wire. The earlier generation of MRI machines mainly detects variations in the concentration and physical characteristics of hydrogen in the body. High field magnets permit the detection of many more elements such as carbon and phosphorous, unlocking new potential for physiological discoveries. The more detailed images provided by these machines could lead to earlier diagnosis and treatment of a variety of diseases including multiple sclerosis and Alzheimer's disease.
The use of stronger magnetic fields increases the speed at which nuclei precess. This, in turn, increases the resonance frequencies, which produce wavelength effects that can result in clearly different distributions of the two circularly polarized components produced by even very simple RF coils. The question about whether only the field component that rotated with precession or both field components were involved in generating an MRI image suddenly became a significant issue in equipment design, particularly in the design of the RF coils. With my experience using XFdtd, electromagnetic simulation software created by Remcom Inc., and active collaborations with researchers who had access to an early 7-Tesla MRI system at the University of Minnesota, I was uniquely positioned to find the answer.
Simulation Goes Beyond Physical Experiments
As an example of our work in MRI safety, Michael Smith of Pennsylvania State University and I used electromagnetic simulation to calculate the RF magnetic field (B1), specific absorption rate (SAR), and SNR as a function of frequency between 64 and 345 MHz for a surface coil against an anatomically accurate human chest. SAR measures tissue exposure to RF fields and is a standard in evaluating MRI safety. The calculated B1 field distributions were in good agreement with previously published experimental results up to 175 MHz, especially considering the dependence of field behavior on subject anatomy. Calculated SNR in the heart agreed well with simple theory for low frequencies. Above 175 MHz, however, the trend in SNR with frequency began to depend largely on location in the heart. At frequencies above 175 MHz, limits on SAR also began to be an issue in some common imaging sequences.
I use Remcom's XFdtd electromagnetic simulation software for my MRI safety studies because it has demonstrated the ability to provide outstanding accuracy in simulation of MRI. XFdtd is used by all three of the leading producers of MRI equipment and its results are considered to be sort of a standard in this type of simulation. XFdtd is based on the finite difference time domain (FDTD) method. According to the Federal Communications Commission of the United States, "Currently, the finite difference time domain (FDTD) algorithm is the most widely accepted computational method for SAR modeling."1
Answering the Question
At 300 MHz there are significant wavelength effects in the spherical sample, resulting in unusual RF field patterns. The results for the RF field components rotating with and opposite nuclear precession are shown respectively in the images marked B1+ and B1- in Figure 3. I compared these distributions to the experimental image, marked SIexp shown in the lower right hand corner of Figure 3. Upon simulating the same situation with XFdtd, we found that neither the circularly polarized magnetic field component rotating with nuclear precession (B1+) nor that rotating opposite nuclear precession (B1-) could be used alone to predict this experimentally acquired image. However, when the distributions of both components were combined in the appropriate way, the resulting simulated image ‘SIcalc' matched the experimental one very well.
The match between SIcalc and SIexp indicates that both components are involved in producing the MR image. Since the time we performed this demonstration, the need for considering the component that rotates opposite nuclear precession has become well accepted and is now an integral part of the prediction of MRI signal for hardware and pulse sequence engineering. This application clearly indicates the value of electromagnetic simulation in designing future generations of MRI equipment. Simulation often enables researchers to investigate applications and effects that cannot be measured experimentally. The result provides insights and understanding that help to improve the design of medical equipment.
Reference1 OET Bulletin 65. "Evaluating Compliance with FCC Guidelines for Human Exposure to Radiofrequency Electromagnetic Fields." Supplement C.
OnlineFor additional information on the technologies and products discussed in this article, see MDT online at www.mdtmag.com or Remcom Inc. at www.remcom.com.