Class I Medical Device Recall: Oridion Medical and Philips Healthcare Microstream CO2 Filterline (FilterLine H Set Infant/Neonate, VitaLine H Set Infant/Neonate)
Recall Class: Class I
Date Recall Initiated: April 7, 2011
Affected models and lots:
|Oridion Medical||006324, 010807||M8330M10,M8386N10,M8411P10,M8451P10,
|Philips Healthcare||M1923A, 989803159581||M8330M10,M8386N10,M8411P10,M8451P10,
The affected devices were manufactured and distributed between October 2010 and February 2011.
The Microstream CO2 Filterline is a medical device used by emergency medical services, hospitals, and other health care providers to measure exhaled carbon dioxide (CO2) during ventilation of newborn and infant patients.
7 Hamarpe St. P.O. box 45025
Jerusalem, Israel 91450
3000 Minuteman Road
Andover, MA 01810
Reason for Recall:
Customers with questions related to this recall should contact Oridion at 1-888-674-3466 ext 1234., Monday – Friday, 8:30 a.m. to 5:30 p.m., Eastern Standard Time.
Customers with questions should contact David Rowe at (978) 659-3021.
FDA District: New England
Philips Healthcare notified all known users of the affected Philips-labeled devices in the U.S. and is in the process of notifying all customers world-wide. In its recall communication, Philips instructed users to immediately identify all products from affected lots, remove them from inventory and dispose in accordance with local regulations.
To date, Oridion Medical and Philips Healthcare have not received any reports of injuries related to this issue.
Class 1 recalls are the most serious type of recall and involve situations in which there is a reasonable probability that use of these products will cause serious adverse health consequences or death.
Health care professionals and consumers may report adverse reactions or quality problems they experienced using these products to MedWatch: The FDA Safety Information and Adverse Event Reporting Program either online, by regular mail or by FAX.